All the abutments, analogues and transfers are made of Grade 5 titanium (Ti6Al4V). The fastening screw fixing ELPSC1 is included in the packaging of the abutments. Tighten to 30 N*cm using the torque ratchet wrench ELPSK1 or ELPSK2 with the torque ratchet ELKWD.
Ball abutment for mobile prosthesis
Tighten to 30 N*cm using the torque ratchet wrench ELSAPK
with the torque ratchet ELKWD.
|EDAN||Gold||Ø 3.5 & Ø 4.0 mm|
|ENAN||Green||Ø 4.75 & Ø 5.25 mm|
|EDTRA||Gold||Ø 3.5 & Ø 4.0 mm|
|ENTRA||Green||Ø 4.75 & Ø 5.25 mm|
Impression transfer technique for the EasyDip® implant
To transfer the location of the fixture onto a casting, proceed by inserting the implant transfer EDTRA or ENTRA in the implant and secure it with the long fastening screw ELPSC2 included in the packaging of the transfer. Tighten using the screwdrivers ELCSK1 or ELCSK2 or the contra-angle wrench ELPSCAK. Then proceed with the impression using a slotted spoon. Before removing the impression, loosen the fastening screw releasing the transfer. Insert the implant analogue EDAN or ENAN, assembling it to its transfer EDTRA using the fastening screw previously loosened. Now it’s possible to send the impression to the laboratory.
Impression transfer technique for ball abutment
To transfer the location of the ball abutments for mobile prosthesis, proceed by screwing abutments themselves with a torque of 30 N*cm, using the torque ratchet wrench ELSAPK inserted into the torque ratchet ELKWD. Take a traditional impression with a traditional spoon and place in the impression the analogues of the ball abutments ELSAPA. Send the impression to the laboratory.